This training set is designed to introduce basic functions of a basic air conditioning unit and to apply basic psychrometric operations.
The training set has a cooling group, resistors for heating operations and a humidification unit. The amount of air used is adjusted by means of an adjustable radial fan. The air channel system has a cross-sectional area of about 0.10 m2. The air sucked from the atmosphere first enters a variable speed centrifugal fan and then through the duct system. During the heating process, the humidification unit is activated to increase the humidity content. In heating process, the heating elements are switched on, the preheating resistance power is 1 kW and the last heating resistance power is 0.5 kW. The air is heated by these resistances. Information from temperature and humidity sensors can be viewed on the LCD screen. If the cooled and nemed air is going to pass through the system, the cooling system will be activated and the air passing through the 3 channels will be cooled by the cooling group and the channel evaporator. As the cool and dry air passes through the duct, the moisture and temperature sensors in this section measure. If air desired to be reheated, the last heating resistors must be switched on. The conditioned air is measured by an air speed meter (anemometer) before the air is released into the atmosphere.
Hermetic reciprocating compressor
Forced air cooled condenser
Aqueous humidifying pads
Digital temperature measurement from 5 different points
Digital relative humidity measurement from 5 different points
Touch LCD Display
USB Computer Connection
Device, device cover, 1 printed experiment report, circuit diagram and product catalog
A x B x H: 1650 x 550 x 1235 mm